Go beyond the standard.®
Canadian industry standards for blood glucose meters have an acceptable error range of 15% from
lab readings. That means that readings on a meter can be 15% higher or lower, compared to the lab.
CONTOUR® NEXT ONE is the most accurate system developed by Ascensia Diabetes Care shown to deliver remarkable accuracy. The CONTOUR® NEXT ONE meter results go beyond the minimum Canadian industry standard for blood glucose meters 2. In a clinical study, the CONTOUR® NEXT ONE meter demonstrated 95% of results were within ±8.4%3 of a lab reading.
1Current regulation based on Health Canada recognized ISO 15197:2013 standard requires results within ±15% range, specifically: ≥95% of results must fall within ±0.83 mmoL/L for blood glucose concentrations <5.55 mmoL/L and within ±15% for blood glucose concentrations ≥5.55 mmoL/L.
2 CONTOUR® NEXT ONE meter meets ±10% accuracy vs. laboratory method, specifically: 97.4% of results within ±10% for blood glucose concentrations ≥5.55 mmol/L, and 100% of results within ±0.56 mmol/L accuracy vs. laboratory method for blood glucose concentrations <5.55 mmol/L.
3In a clinical study, conducted in accordance with the requirements of ISO 15197:2013, the CONTOUR® NEXT ONE not only met and exceeded the minimum requirements of the standard but an ad hoc analysis determined where 95% of results fell during the study, which was within ±0.47 mmol/L or ±8.4% of the laboratory reference value for glucose concentrations <5.55 mmoL/L or ≥ 5.55 mmoL/L, respectively, when tested via subject obtained capillary fingertip results (patients). Christiansen M et al. J Diabetes Sci Technol 2017;11(3):567-573.